- Research article
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
Correlation between penile cuff test and pressure-flow study in patients candidates for trans-urethral resection of prostate
© Bianchi et al.; licensee BioMed Central. 2014
Received: 25 September 2014
Accepted: 11 December 2014
Published: 19 December 2014
Aim of this study was to make a comparison between penile cuff test (PCT) and standard pressure-flow study (PFS) in the preoperative evaluation of patients candidates for trans-urethral resection of prostate (TURP) for benign prostatic obstruction (BPO).
We enrolled male patients with lower urinary tract symptoms candidates for TURP. Each of them underwent a PCT and a subsequent PFS. A statistical analysis was performed: sensitivity (SE), specificity (SP), positive predictive value (PPV), negative predictive value (NPV), likelihood ratio and ratio of corrected classified were calculated. Fisher exact test was used to evaluate relationships between PCT and maximal urine flow (Q max): a p- value < 0.05 was considered statistically significant.
We enrolled 48 consecutive patients. Overall, at PCT 31 patients were diagnosed as obstructed and 17 patients as unobstructed. At the subsequent PFS, 21 out of 31 patients diagnosed as obstructed at PCT were confirmed to be obstructed; one was diagnosed as unobstructed; the remaining 9 patients appeared as equivocal. Concerning the 17 patients unobstructed at PCT, all of them were confirmed not to be obstructed at PFS, with 10 equivocal and 7 unobstructed. The rate of correctly classified patients at PCT was 79% (95%-CI 65%-90%). About detecting obstructed patients, PCT showed a SE of 100% and a SP of 63%. The PPV was 68%, while the NPV was 100%.
PCT can be an efficient tool in evaluating patients candidates for TURP. In particular, it showed good reliability in ruling out BPO because of its high NPV, with a high rate of correctly classified patients overall. Further studies on a huger number of patients are needed, including post-operative follow-up as well.
In clinical practice, when required, a proper evaluation and quantification of BPO is performed by invasive UD, in particular pressure-flow study (PFS) .
Over the last two decades, some alternative, less invasive tests have been proposed , based on equipment consisting of an external condom catheter , an intra-urethral device  or an inflatable cuff around the penis – penile cuff test (PCT) – with inflation-deflation cycles .
Instead of the direct intravesical sampling used in PFS, non-invasive UD aims to give information about bladder pressure by evaluating the equal urine pressure either along the urethra (in penile cuff), or at the external meatus (in external condom catheter).
PCT results can be plotted on the nomogram proposed by Griffiths  which is designed on a cartesian plane with maximal urine flow (Q max) on the x-axis and p cuff on y-axis, with an ascending straight line, with y-intercept equal to 80 cm H 2 O, separating obstructed from non-obstructed patients.
Recently, a new prototype of PCT has been proposed , using an automatically controlled inflatable cuff which detects bladder voiding pressure at constant low urine flow instead of inflation-deflation cycles.
The purpose of this study was to compare the data of PCT with inflation-deflation cycles with those of a standard PFS in patients candidates for trans-urethral resection of prostate (TURP).
Male patients who previously received indication to undergo a TURP in our or in a different center were included. Indication for TURP had been made on referral urologist opinion, generally on the basis of the presence of lower urinary tract symptoms (LUTS) and a reduced flow rate, independently by other aspects as prostate volume and alpha-blockers effectiveness.
A urine sample for urine culture was collected by spontaneous micturition within 7 days before the tests in order to rule out possible infections. Exclusion criteria were diabetes mellitus, any neurological disease, use of drugs impairing bladder contractility or impacting on lower urinary tract function, an indwelling bladder catheter over the previous six months, presence of urinary tract infection, suspect of malignancies.
Approval of the study by Ethics Committee of Policlinico Tor Vergata was obtained, and all patients signed a written informed consent to be included. For each patient, we performed a PCT followed by a subsequent PFS and both procedures were conducted by the same urodynamicist.
PCT was performed by Mediplus CT3000 Cuff Machine®, which allows multiple inflation-deflation cycles during a single micturition, getting several BIP measurements.
The patients were instructed to perform a micturition without abdominal straining.
For each inflation cycle we applied the exclusion rules proposed by Drinnan et al. , thus a cycle was immediately excluded in case of one of the following conditions:
No flow recovery after cuff deflation, meaning that the micturition ended during the last cycle, so the cuff pressure could have been not responsible for urine flow stop;
There was an ‘erratic’ flow trace, which could be related to straining or maybe to contractions by the pelvic floor or membranous urethra;
The urine flow was not interrupted at the device maximum pressure, which is set at 200 cm H 2 O for safety reasons. This situation is associated with highly contractile bladder.
Furthermore, we repeated any test showing a total bladder volume less than 150 mL.
PFS was performed by a urodynamic equipment (Life-Tech®, Stafford, TX, USA) with water-filled bladder catheter and rectal balloon (Life-Tech®, Stafford, TX, USA), after a filling phase with non-physiological filling rate (30–50 mL/s).
For each of two categories – obstructed versus non-obstructed – patients were subdivided into two subgroups according to their Q max, with a threshold of 10 mL/s , in order to evaluate if Q max was able to improve accuracy.
Sensitivity (SE), specificity (SP), positive predictive value (PPV), negative predictive value (NPV), likelihood ratio and ratio of corrected classified were calculated.
In order to assess accuracy of estimated value, 95% confidence interval (95%-CI) was calculated for SE, SP, PPV, NPV and ratio of corrected classified.
Fisher exact test was used to evaluate relationships between PCT and Q max: a p- value < 0.05 was considered statistically significant.
Software Stata 13.0 (College Station®, Texas) was used for all analysis.
Results have been reported according to Standards for Reporting of Diagnostic accuracy (STARD) flow-chart .
The research was carried out in compliance with the Helsinki Declaration - Ethical Principles for Medical Research Involving Human Subjects.
We enrolled 48 consecutive male patients – mean age 61.5 ± 13.1 years.
On uroflowmetry, median Q max was 11.6 mL/s (range 4.0-25.0 mL/s), with a median post-void residual urine volume (PVR) of 42 mL (range 0–430 mL) detected by ultrasound scan. PSA mean value was 2.35 ng/mL (range 0.80-4.10 ng/mL).
No adverse events occurred during or after the tests.
Overall, at PCT 31 patients were diagnosed as obstructed and 17 patients as unobstructed on Griffiths nomogram .
On the subsequent PFS, according to Abrams-Griffiths nomogram [12, 13], 21 out of 31 patients diagnosed as obstructed at PCT were confirmed to be obstructed; one was diagnosed as unobstructed; the remaining 9 patients appeared as equivocal.
Results of penile cuff test (PCT) compared with pressure-flow studies (PFS)
PENILE CUFF obstructed
PENILE CUFF unobstructed
The rate of correctly classified patients at PCT was 79% (95%-CI 65%-90%).
About pressure measurements, overall at PCT we obtained a mean p cuff equal to 133.75 cm H 2 O (SD 33.45 cm H 2 O), while at PFS we had a mean detrusorial pressure at Q max flow equal to 52.69 cm H 2 O (SD 21.94 cm H 2 O).
Focusing on obstructed patients, at PCT we had a mean p cuff =157.00 cm H 2 O (SD 26.83 cm H 2 O), while at PFS the mean detrusorial pressure was 74.00 cm H 2 O (SD 14.13 cm H 2 O).
With regard to the further subdivision according to Q max, 15 patients out of the 31 obstructed at PCT showed a Q max < 10 mL/s, with the other 16 patients having a Q max ≥ 10 mL/s.
On the other hand, among the 17 unobstructed patients on PCT, we had 5 with a Q max < 10 mL/s and 12 with a Q max ≥ 10 mL/s.
Patients categorization into subgroups according to their Q max greater or less than 10 mL/s did not produce a further improvement of PCT ability to diagnose BPO (Fisher exact test, p = 0.2362).
Results of penile cuff test (PCT) compared with pressure-flow studies (PFS)
PFS O (n)
PFS U (n)
PFS O (n)
PFS U (n)
LR + %
Over the last century, a simple evaluation of PVR has been proposed as an appealing tool for diagnosing BPO. Nevertheless, a huge PVR may be due to an impaired detrusor contractility (IDC) . In fact, it has been pointed out that up to one half of unobstructed patients with LUTS could have elevated PVR, while up to one forth of severely obstructed patients could show no PVR .
Thus, the association between elevated PVR and BPO is not strong enough to be used as a useful clinical tool .
In some papers, the role of uroflow trace has been investigated as well, but no reliable relation between its profile and BPO was found out .
Some Authors have shown that uroflowmetry could be able to assess the presence of BPO in the vaste majority of patients with Q max less than 10 mL/s, with a progressively decreasing rate of BPO in case of Qmax major than 10 mL/s [1, 2, 14].
Accordingly, European Association of Urology guidelines have considered PFS as an optional test before surgery for BPO, usually indicated in the preoperative evaluation of patients showing a Q max > 15 mL/s .
Conversely, 25-30% of men with decreased Q max at uroflowmetry are unobstructed . Indeed, decreased uroflow can result from either impaired detrusor contractility (IDC) or BPO; thus, only detrusor pressure measurement is able to distinguish between those conditions [18, 20].
As a consequence, PFS still represents the gold standard for a proper evaluation of BPO in male patients, above all when an IDC is suspected .
In clinical practice, the nomogram proposed by Abrams and Griffiths for the diagnosis of obstruction in males at PFS has been widely used .
A further nomogram proposed by Abrams  aims to give a more accurate patients categorization by the introduction of bladder outlet obstruction index (BOOI) and bladder contractility index (BCI).
Thus, PFS represents the gold standard for the evaluation of BPO . Nevertheless, this test is not used as a routine examination before surgery for BPO , because it is considered time-consuming, not cost-effective overall  and a potential cause of morbidity .
Over the last 30 years, no simple tool proved to be reliable in distinguishing between BPO and IDC.
Aim of our study was to make a comparison between PCT and PFS in the diagnostic work-up on patients candidates for TURP: summarizing our results, PCT showed a SE of 100% and a SP of 63% in detecting obstructed patients, with a PPV of 68% and a NPV of 100%.
Overall, the rate of correctly classified patients at PCT was 79%. In particular, non-obstructed patients at PCT were confirmed as non-obstructed at PFS.
Using the nomogram modified for non-invasive pressure measurement, Griffiths et al.  obtained with PCT a PPV of 68% and a NPV of 78% for PFS diagnosis of BPO. Besides, they noticed that predictive accuracy for obstruction could be improved by the additional criterion of Q max less than 10 mL/s, thus obtaining a PPV of 88% and a NPV of 86%.
In our study, patients categorization into subgroups according to their Q max (threshold 10 mL/s) did not add any further information as it did not get confirmation of its statistical significance at Fisher exact test.
The difference between Griffiths’ and our results could be due to a different selection of patients: in our study, only patients who were candidates for a TURP  were enrolled, while in Griffiths’ paper the Authors intended to analyze patients complaining with LUTS . Nevertheless, further studies are needed to investigate this aspect.
Our data seem to confirm those ones obtained in another, more recent paper based on 30 consecutive patients complaining with LUTS. In this study, Borrini L et al.  found for PCT a PPV of 82% and a NPV of 88% for BPO at PFS.
According to our experience, non-invasive UD, in particular PCT, can be a useful diagnostic tool in patients candidates for TURP, suggesting a possible solution to the thorny problem about urodynamic tests before surgery for BPO. In fact, compared to PFS, PCT appeared as a quick and accurate test to rule out a BPO condition because of its high NPV. Thus it could be used to run a selection of non-obstructed patients suspected for an eventual IDC condition. Indeed, such patients are the most critical ones in BPO surgery , with some papers reporting about one forth of them showing no symptoms improvement after a surgical treatment [24, 26].
In a paper by Harding et al. , a consecutive cohort of 208 men undergoing TURP were previously evaluated by PCT: 87% of patients diagnosed with BPO had a clinical improvement after surgery, while only 56% of patients deemed as not obstructed had a good outcome.
By such diagnostic pathway, non-obstructed patients (probably with IDC) could be easily recognized and adequately counseled in advance about the prospect of poor or partial symptoms improvement after surgery for BPO, avoiding a PFS.
Furthermore, the rate of correctly classified patients at PCT was high, confirming that most obstructed patients can be adequately evaluated by PCT.
We should also consider that PCT categorization does not allow for ‘equivocal’ patients, who finally represent the mismatch between the two urodynamic tests. Anyway, those patients can be mostly considered eligible to surgery, so PFS could be neglected in such cases.
Only patients with an unclear diagnosis could be suggested to undergo PFS, while the other ones could be probably evaluated just by PCT, getting the amount of pre-operative information useful both to the surgeon and to the patient, in terms of preoperative counseling.
Limitations of our study are the relatively small sample size and the lack of a post-operative follow-up to assess TURP efficacy in different categories of patients.
On the other hand, this is, to our knowledge, the first study comparing PCT to PFS in patients candidates for TURP.
Further papers on large series of patients including post-operative follow-up are needed, in order to assess the real role of PCT in the pre-operative evaluation for BPO.
PCT can be an efficient tool in evaluating patients candidates for TURP. In particular, it showed good reliability in ruling out BPO because of its high NPV, with a high rate of correctly classified patients overall. Further studies based on a bigger sample size are needed, including post-operative follow-up.
- Nitti VW: Pressure Flow Urodynamic Studies: The Gold Standard for Diagnosing Bladder outlet Obstruction. Rev Urol. 2005, 7 (Suppl 6): S14-S21.PubMedPubMed CentralGoogle Scholar
- Mangera A, Osman NI, Chapple CR: Assessment of BPH/BOO. Indian J Urol. 2014, 30 (2): 177-180. 10.4103/0970-1591.126902. doi:10.4103/0970-1591.126902View ArticlePubMedPubMed CentralGoogle Scholar
- Blake C, Abrams P: Noninvasive techniques for the measurement of isovolumetric bladder pressure. J Urol. 2004, 171 (1): 12-19. 10.1097/01.ju.0000102685.44036.b9.View ArticlePubMedGoogle Scholar
- Huang Foen Chung JW, Bohnen AM, Pel JJ, Bosch JL, Niesing R, van Mastrigt R: Applicability and reproducibility of condom catheter method for measuring isovolumetric bladder pressure. Urology. 2004, 63: 56-60. 10.1016/j.urology.2003.08.030.View ArticlePubMedGoogle Scholar
- D’Ancona CAL, Bassani JWM, de Oliveira Querne FA, Carvalho J, Oliveira RR, Netto NR: New method for minimally invasive urodynamic assessment in men with lower urinary tract symptoms. Urology. 2008, 71: 75-78. 10.1016/j.urology.2007.08.036.View ArticlePubMedGoogle Scholar
- Griffiths CJ, Rix D, MacDonald AM: Noninvasive measurement of bladder pressure by controlled inflation of a penile cuff. J Urol. 2002, 167 (3): 1344-1347. 10.1016/S0022-5347(05)65296-2.View ArticlePubMedGoogle Scholar
- Griffiths CJ, Harding C, Blake C, McIntosh S, Drinnan MJ, Robson WA, Abrams P, Ramsden PD, Pickard RS: A nomogram to classify men with lower urinary tract symptoms using urine flow and noninvasive measurement of bladder pressure. J Urol. 2005, 174 (4 Pt 1): 1323-1326.View ArticlePubMedGoogle Scholar
- Clarkson B, Griffiths C, McArdle F: Continuous non-invasive measurement of bladder voiding pressure using an experimental constant low-flow test. Neurourol Urodyn. 2012, 31: 557-563. 10.1002/nau.21170.View ArticlePubMedGoogle Scholar
- Drinnan MJ, McIntosh SL, Robson WA, Pickard RS, Ramsden PD, Griffiths CJ: Inter-observer agreement in the estimation of bladder pressure using a penile cuff. Neurourol Urodyn. 2003, 22: 296-300. 10.1002/nau.10133.View ArticlePubMedGoogle Scholar
- Schafer W, Abrams P, Liao L, Mattiasson A, Pesce F, Spangberg A, Sterling AM, Zinner NR, van Kerrebroeck P, International Continence Society: Good urodynamic practices: uroflowmetry, filling cystometry and pressure-flow studies. Neurourol Urodyn. 2002, 21 (3): 261-274. 10.1002/nau.10066.View ArticlePubMedGoogle Scholar
- Abrams P, Cardozo L, Fall M, Griffiths D, Rosier P, Ulmsten U, van Kerrebroeck P, Victor A, Wein A, Subcommittee of the International Continence Society: The standardisation of terminology of lower urinary tract function: report from the Standardistion Sub-committee of the International Continence Society. Neurourol Urodyn. 2002, 21: 167-178. 10.1002/nau.10052.View ArticlePubMedGoogle Scholar
- Griffiths D, Hofner K, van Mastrigt R, Rollema HJ, Spangberg A, Gleason D: Standardisation of terminology of lower urinary tract function: pressure flow studies of voiding, urethral resistance and urethral obstruction. Neurourol Urodyn. 1997, 6: 1-18.View ArticleGoogle Scholar
- Abrams P: Bladder outlet obstruction index, bladder contractility index and bladder voiding efficiency: three simple indices to define bladder voiding function. BJU Int. 1999, 84: 14-15. 10.1046/j.1464-410x.1999.00121.x.View ArticlePubMedGoogle Scholar
- Reynard JM, Yang Q, Donovan JL, Peters TJ, Schafer W, de la Rosette JJ, Dabhoiwala NF, Osawa D, Lim AT, Abrams P: The ICS-'BPH' Study: uroflowmetry, lower urinary tract symptoms and bladder outlet obstruction. Br J Urol. 1998, 82 (5): 619-623. 10.1046/j.1464-410X.1998.00813.x.View ArticlePubMedGoogle Scholar
- Bossuyt PM, Reitsma JB, Bruns DE, Gatsonis CA, Glasziou PP, Irwig LM, Moher D, Rennie D, de Vet HC, Lijmer JG: Standards for Reporting of Diagnostic accuracy. The STARD statement for reporting studies of diagnostic accuracy: explanation and elaboration. Ann Intern Med. 2003, 138 (1): W1-W12. 10.7326/0003-4819-138-1-200301070-00012-w1.View ArticlePubMedGoogle Scholar
- Abrams PH, Griffiths D: The assessment of prostatic obstruction from urodynamic measurements and from residual urine. BJU. 1979, 51: 129-134.b. 10.1111/j.1464-410X.1979.tb02846.x.View ArticlePubMedGoogle Scholar
- Griffiths DJ: Pressure-flow studies of micturition. Urol Clin North Am. 1996, 23: 279-297. 10.1016/S0094-0143(05)70311-6.View ArticlePubMedGoogle Scholar
- Chancellor MB, Blaivas JG, Kaplan SA, Axelrod S: Bladder outlet obstruction versus impaired detrusor contractility: the role of outflow. J Urol. 1991, 145 (4): 810-812.PubMedGoogle Scholar
- Oelke M, Bachmann A, Descazeaud A, Emberton M, Gravas S, Michel MC, N'dow J, Nordling J, de la Rosette JJ: Guidelines on the management of male lower urinary tract symptoms, incl. benign prostatic obstruction. Eur Assoc Urol. 2013, 8-9.Google Scholar
- Gerstenberg TC, Andersen JT, Klarskov P, Ramirez D, Hald T: High flow infravesical obstruction in men: symptomatology, urodynamics and the results of surgery. J Urol. 1982, 127: 943-945.PubMedGoogle Scholar
- Porru D, Madeddu G, Campus G, Montisci I, Scarpa RM, Usai E: Evaluation of morbidity of multi-channel pressure-flow studies. Neurourol Urodyn. 1999, 18 (6): 647-52.22. 10.1002/(SICI)1520-6777(1999)18:6<647::AID-NAU15>3.0.CO;2-N.View ArticlePubMedGoogle Scholar
- Borrini L, Lukacs B, Ciofu C, Gaibisso B, Haab F, Amarenco G: Predictive value of the penile cuff-test for the assessment of bladder outlet obstruction in men. Prog Urol. 2012, 22 (11): 657-664. 10.1016/j.purol.2012.07.017.View ArticlePubMedGoogle Scholar
- Arnolds M, Oelke M: Positioning invasive versus non invasive urodynamics in the assessment of bladder outlet obstruction. Curr Opin Urol. 2009, 19 (1): 55-62. 10.1097/MOU.0b013e328317ca9b.View ArticlePubMedGoogle Scholar
- Neal DE, Ramsden PD, Sharples L, Smith A, Powell PH, Styles RA, Webb RJ: Outcome of elective prostatectomy. BMJ. 1989, 299 (6702): 762-767. 10.1136/bmj.299.6702.762.View ArticlePubMedPubMed CentralGoogle Scholar
- Losco G, Keedle L, King Q: Non-invasive urodynamics predicts outcome prior to surgery for prostatic obstruction. BJU Int. 2013, 112 (Suppl 2): 61-64.View ArticlePubMedGoogle Scholar
- Emberton M, Neal DE, Black N: The effect of prostatectomy on symptom severity and quality of life. Br J Urol. 1996, 77 (2): 233-247. 10.1046/j.1464-410X.1996.88213.x.View ArticlePubMedGoogle Scholar
- Harding C, Robson W, Drinnan M, Sajeel M, Ramsden P, Griffiths C, Pickard R: Predicting the outcome of prostatectomy using non-invasive bladder pressure and urine flow measurements. Eur Urol. 2007, 52: 186-192. 10.1016/j.eururo.2006.11.009.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2490/14/103/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.